For more info: 021 55734344
FEATURES:
 Wide Test Frequency 20Hz ~ 10/5/1MHz
 0.1% Basic Accuracy & 6 Digits Measurement Resolution
 Large LCD Display with Intuitive Interface
 Full Measuring Functions with DUT V/I Monitor
 PASS/FAIL Function ( abs, % or delta) with Judgment Alarm
 DC Resistance Measurement
 Multiple Step Mode
 Standard RS232C/GPIB Interface
 Optional DC Bias box
The LCR8000G Series LCR meter, with test frequency up to 10MHz, provides accuracy, versatility and high resolution for a wide range of component measurements, even including DC resistance measurement and Voltage/Current monitoring. The MultiStep function allows onscreen programming of customized measurement sequence with Pass/Fail indication. Each program includes 30 test steps and each test step can be set with selected parameters and test limits. Under MultiStep operation, a tedious work routine can be done step by step automatically just at a press of a button. With Graph Mode, LCR8110G, LCR8105G and LCR8101G display the component impedance response either over a wide range of test frequency sweep or over a wide range of test voltage sweep in a graph chart. This gives an analysis result of either impedance vs. frequency or impedance vs. applied voltage all at a glance. GPIB and RS232C interfaces are available as standard for instrument control and test result display on the PC. The rich features of LCR8000G Series easily make your measurement tasks done at a very competitive price.
TEST FREQUENCY  
20Hz ~ 10 / 5 / 1 MHz, 5 Digits, ±0.005%  
INPUT IMPEDANCE  
100Ω  
BASIC ACCURACY (*)  
±0.1% (R, Z, X, G, Y, B, L, C)  
TEST SPEED  
AC (＞ 2kHz) MAX: 75mS FAST: 150mS MEDIUM: 450mS SLOW: 600mS 
DC MAX: 30mS FAST: 60mS MEDIUM: 120mS SLOW: 900mS 
TEST SIGNAL LEVELS  
≦3MHz：10mV~2Vrms, 1mV/Step or 10mV/Step, 2%±5mV ＞3MHz：10mV~1Vrms, 1mV/Step or 10mV/Step, 2%±5mV 

SHORT CIRCUIT CURRENT  
Max. 20mA  
MEASUREMENT RANGES  
R, Z, X  0.1mΩ ~ 100MΩ 
G, Y, B  10nS ~ 1000S 
L  0.1nH ~ 100kH 
C  0.01pF ~ 1F 
D  0.00001 ~ 9.9999 
Q  0.1 ~ 9999.9 
θ  180° ~ +180° 
Rdc  0.1mΩ ~ 100MΩ 
MEASUREMENT PARAMETERS  
Impedance (Z), Phase Angle (θ), Inductance (L), Capacitance (C), AC Resistance (Rac), Quality Factor (Q), Dissipation Factor (D), Admittance (Y), Conductance (G), Reactance (X), Susceptance (B), DC Resistance (Rdc)  
SERIES OR PARALLEL EQUIVALENT CIRCUIT  
C + R, C + D, C + Q, L + R, L + Q, L + D  
SERIES EQUIVALENT CIRCUIT ONLY  
X + R, X + D, X + Q  
PARALLEL EQUIVALENT CIRCUIT ONLY  
C + G, B + G, B + D, B + Q, B + R, L + G  
POLAR FORM  
Z + Phase Angle, Y + Phase Angle  
AVERAGE  
1 ~ 256 TIMES  
LCD DISPLAY  
320 x 240 DOTMATRIX  
INTERFACE  
RS232, GPIB  
POWER SOURCE  
AC 115V(+10% / 25%), 230V(+15% / 14%) (Selectable), 50/60Hz  
DIMENSIONS & WEIGHT  
330(W) x 170(H) x 340(D)mm, Approx. 5kg 
(*)：Basic accuracy varies with the speed, frequency, AC signal level and impedance of the device under test.