For more info: 021 55734344
In the refrigeration circuit the expansion valves, also called primary controllers, also play an important role. The correct selection and adjustment decisively affects the capacity of the overall refrigeration system.
The trainer enables the investigation of four different expansion valves: TEV (Thermostatic Expansion Valve) with internal pressure compensation, TEV with external pressure compensation, TEV with MOP function (Maximum Operating Pressure) and AEV (Automatic Expansion Valve). The expansion valves can be selected via valves.
Via an evaporator with glass tubes the correct injection and evaporation of the refrigerant can be monitored. An additional electrically heated superheater at the output of the evaporator enables the investigation of the control behaviour with different refrigerant superheating. The locations of the temperature sensor at the evaporator output can be selected. The control behaviour can be monitored at a flow meter. Pressure and temperature sensors upstream and downstream of the expansion valve and at the evaporator outlet provide information about the state of the refrigerant and the degree of superheating.
A commercial condensing unit completes the refrigeration circuit.
The measured values can be read on digital displays. At the same time, the measured values can also be transmitted directly to a PC via USB. The data acquisition software is included. In the software the control behaviour of the expansion valves is recorded over time. The trainee can check the correct adjustment of the expansion valve and investigate effects such as hunting. The well-structured instructional material sets out the fundamentals and provides a step-by-step guide through the exercises.
Learning Objectives / Experiments
- operation and properties of the various expansion valves
- investigation of the static control characteristic
- investigation of the dynamic control behaviour
- adjustment of the optimum operating point
- benefits and disadvantages of the different expansion valves
* thermostatic expansion valve (TEV) with internal pressure compensation
* TEV with external pressure compensation
* TEV with MOP function (Max. Operating Pressure)
* pressure-controlled expansion valve (AEV)
- hunting effect
 trainer from the GUNT practical series for the training of mechatronics engineers for refrigeration
 refrigeration system with different expansion elements: thermostatic expansion valve TEV with internal pressure compensation, thermostatic expansion valve TEV with external pressure compensation, thermostatic expansion valve TEV with MOP function, pressure-controlled expansion valve AEV
 evaporator with glass tubes to monitor injection and evaporation
 valve downstream of the evaporator to simulate different pressure losses
 adjustable electric heater for superheating
 air-cooled condensing unit
 sensors record pressures, temperatures, flow rate
 refrigerant R134a, CFC-free
 GUNT software for data acquisition with USB under Windows Vista or Windows 7
Air-cooled condensing unit
- power consumption: 485W at -10/32°C
- refrigeration capacity: 675W at -10/32°C
- receiver: 1,4L
- pressure: 2x -1...9bar, 1x -1...24bar
- temperature: 4 x -40...150°C
- flow rate: 3...41L/h
Dimensions and Weight
Weight: approx. 130kg Required for Operation 230V, 50Hz, 1 phase
Scope of Delivery
1 GUNT software CD + USB cable
1 set of instructional material